When an interface exists between a liquid and a solid, the angle between the surface of the liquid and the outline of the contact surface is described as the contact angle θ (lower case theta). The contact angle (wetting angle) is a measure of the wettability of a solid by a liquid.
In the case of complete wetting (spreading), the contact angle is 0°. Between 0° and 90°, the solid is wettable and above 90° it is not wettable. In the case of ultrahydrophobic materials with the so-called lotus effect, the contact angle approaches the theoretical limit of 180°.
Schematic diagram of contact angle
The contact angle with one or more liquids can be used to determine the surface free energy of a solid.
The contact angle is important wherever the intensity of the phase contact between liquid and solid substances needs to be checked or assessed: coating, painting, cleaning, printing, hydrophobic or hydrophilic coating, bonding, dispersing etc.
- Drop shape analysis (DSA): The contact angle is measured using the image of a sessile drop at the points of intersection (three-phase contact points) between the drop contour and the projection of the surface (baseline). Instruments: DSA100, DSA30, DSA25, MSA, MobileDrop
- Wilhelmy plate method: The force acting in the tensile direction when moving a plate-shaped solid vertically in a liquid is measured. This force depends on the contact angle as well as on the surface tension and the wetted length. Instrument: K100
- Powder contact angle measurement using the Washburn method: The increase in weight of an immersed powder-filled tube is measured with respect to time. The rate of rise of the liquid column depends, among other things, on the contact angle. Instrument: K100
- Top-view distance method: The curvature of the drop surface associated with the contact angle is measured using the distance between light spots which are reflected on the top of a drop surface. Instrument: TVA100
- Adsorption coefficient
- Advancing angle
- Bubble pressure tensiometer
- Captive bubble method
- Circle method
- Conic section method
- Contact angle
- Critical surface tension
- Diffusion coefficient
- Dispersive part
- Drop shape analysis
- Drop volume tensiometer
- Du Noüy ring method
- Dynamic contact angle
- Dynamic surface tension
- Equation of state
- Extended Fowkes method